The U.S. Food and Drug Administration (FDA) today allowed marketing of the DEKA Arm System, the first prosthetic arm that can perform multiple, simultaneous powered movements controlled by electrical signals from electromyogram (EMG) electrodes.
Deka calls it "Luke", after Luke Skywalker of "Star Wars" fame, and the project was funded by DARPA.
EMG electrodes detect electrical activity caused by the contraction of muscles close to where the prosthesis is attached. The electrodes send the electrical signals to a computer processor in the prosthesis that translates them to a specific movement or movements.
The EMG electrodes in the DEKA Arm System convert electrical signals into up to 10 powered movements, and it is the same shape and weight as an adult arm. In addition to the EMG electrodes, the DEKA Arm System contains a combination of mechanisms including switches, movement sensors, and force sensors that cause the prosthesis to move.
Credit: Deka Research
“This innovative prosthesis provides a new option for people with certain kinds of arm amputations,” said Christy Foreman, director of the Office of Device Evaluation at the FDA’s Center for Devices and Radiological Health. “The DEKA Arm System may allow some people to perform more complex tasks than they can with current prostheses in a way that more closely resembles the natural motion of the arm.”
The FDA reviewed clinical information relating to the device, including a 4-site Department of Veterans Affairs study in which 36 DEKA Arm System study participants provided data on how the arm performed in common household and self-care tasks. The study found that approximately 90 percent of study participants were able to perform activities with the DEKA Arm System that they were not able to perform with their current prosthesis, such as using keys and locks, preparing food, feeding oneself, using zippers, and brushing and combing hair.
The DEKA Arm System can be configured for people with limb loss occurring at the shoulder joint, mid-upper arm, or mid-lower arm. It cannot be configured for limb loss at the elbow or wrist joint.
Data reviewed by the FDA also included testing of software and electrical and battery systems, mitigations to prevent or stop unintended movements of the arm and hand mechanisms, durability testing (such as ability to withstand exposure to common environmental factors such as dust and light rain), and impact testing.
The FDA reviewed the DEKA Arm System through its de novo classification process, a regulatory pathway for some novel low- to moderate-risk medical devices that are first-of-a-kind.
The DEKA Arm System is manufactured by DEKA Integrated Solutions in Manchester, N.H.
- PHYSICAL SCIENCES
- EARTH SCIENCES
- LIFE SCIENCES
- SOCIAL SCIENCES
Subscribe to the newsletter
Stay in touch with the scientific world!
Know Science And Want To Write?
- Intellectually Gifted Kids And Learning Disabilities Often Go Hand In Hand
- Why It Took Big Humans To Populate Europe
- Another One Bites The Dust - WW Cross Section Gets Back Where It Belongs
- ESA’s IXV spaceplane Test Flight a Complete Success
- When Does Quantum Mechanics Become Classical Physics?
- Spring Flukes: New 3-Sigma Signals From LHCb And ATLAS
- "Who are we kidding our education systems (worldwide) are designed for the average child? Yes there..."
- "Thanks for the post to let us know more about herpes...."
- " Fermi's Paradox seems to be assuming advanced civilizations would be biologic lifeforms instead..."
- "Ginger ale may be as close to salty as I can go...."
- "Hi Hank - lots of good comments - some I agree with - others not so much, LOL. The old saying of..."